Weighing sensor is applied in the real life most of a sensor, a lot of ordinary people can be seen in the home, today we say products in industrial application. Weighing sensor itself has the advantages of firm structure, durable, but if you don’t pay attention to in the use of the installation process, may also affect its performance.
For small measurement sensor, should avoid it, it will affect the accuracy of measurement value of it in the future. Important heavy large dose of the product itself, it is best to use when handling special handling tools, fell to the ground to slow speed, avoid vibration damages to the it devices, installed base is placed at the same time, to clean, at the bottom of the clutter can’t exist. Any weighing sensor in use process need to avoid overload measurement, which can lead to elastic body deformation irreversible, causing permanent damage.
Sensors in the process of level adjustment, it is best to use level, makes the installation of sensors in the horizontal plane, to ensure the consistency by the load.
Sensors to adopt bolt, the guarantee of fastening torque, long time use would not loosen.
Sensor installation position as far as possible away from high power devices, because its output signal is weak, usually for millivolt level, susceptible to interference. In this case, the sensor cap, output signal selection with shielded twisted-pair cable transmission, its performance will be better. The cable laying and avoid and parallel cable. Must be placed parallel, the distance between them must be greater than 50 centimeters, signal cable with a professional casing protection at the same time, the shield wire must be single measuring grounding and not with the power cord.
Installed in the weighing sensor should pay attention to rainproof outdoor radiation protection.
For a long sensor has invested time, can be used to add a small load, to check whether there is change, the output of its moving parts may be the influence of clutter, reduce its output precision.